Completely stand-alone unit after initial configuration of test parameters
MCB Range Selection : Through neatly labelled shorting links
Chart provided on the panel for using the right CT links and Voltage Taps for required test currents (1.13 In and 1.45 In
Instantaneous Bypass arrangement on each station in the event of MCB trip to maintain uninterrupted current
Arrangement to connect lugs of various sizes inside the heating chamber
Individual test time measurement for each station
Graded inductors to maintain required OCV at the source output (Min 30 V as per standard) especially while testing less number of MCBs
Change of test current (from 1.13 In to 1.45 In) after set time (usually 1 hour) is automatic.
The user may simultaneously test MCBs of different brands and compare the test results (Ideal Application for Test Labs and Certification Agencies).
PC Based Option Available (Extra) for control and data keeping on a PC (SCADA) with customized software
User friendly digital interface to input test parameters
2.55 test on reduced number of stations by user setting